ME ist umgezogen!
Ab dem 12.10.2020 finden Sie uns im ME-Firmengebäude.
Die neue Adresse lautet:
Eduard-Maurer Str. 9
This video instruction explains how to connect the measuring amplifier GSV with the software GSVmulti and how to set the digital display (scaling) as well as the measuring frequency.
This video instruction explains how to connect the measuring amplifier GSV with the software GSVmulti and how to apply GSVmulti with 6-axis sensor.
-loading .tdms file with measuring dataand reading, plotting and exporting to Excel
This video shows how to save and to load settings of sensor (as scaling, measurement frequency, device settings and channel names) and explains which settings are saved in a .ucf file and which settings are kept in the internal memory.
This video gives a brief overview of the special features of the GSV-8.
This video shows how the digital inputs and outputs of the GSV-8 measuring amplifier are configured.
This video shows how the data format of the GSV-8 measuring amplifier can be changed from "Float" to 24Bit or 16Bit. In the format "Float" 8000 measured values (sum sampling rate 64 kHz) can be measured via USB 8x. In the formats "24Bit" and "16Bit", 24000 measured values (sum sampling rate 192 kHz) can be transmitted via USB 8x.
This video shows how the analog output of the GSV-8 is configured: gain, offset, and output type 4 ... 20mA or + -10V ....) can be configured.