Wir stellen aus!
01. - 05. April 2019 in Hannover.
Wir freuen uns auf Ihren Besuch in der Halle 11, Stand F31.
We are looking forward to your visit in Hall 11, stand F31.
Weitere Messen / More trade fairs
Via the menu bar, you can get from "Device" -> "Advanced Settings" -> "Value Mode" to the setting of the data format and the transmission behavior.
The default data format of the GSV-8 is "Float".
By selecting 24 bits, a data frequency can be set of up to 24000 Hz.
By selecting 16Bit you can set a maximum data frequency of 48000Hz.
After switching on the GSV-8, it continuously transmits measured values with the set data frequency. (Values transmitted permanently)
By selecting "Values NOT transmitted permanently (" Logger ") the GSV-8 sends individual measured values only on request (for example by the command" Getvalue ", or by hardware trigger.
The ASCII format and the "Logger" mode can be set in particular for the GSV-2 and GSV-3 series amplifiers.