ME ist umgezogen!
Ab dem 12.10.2020 finden Sie uns im ME-Firmengebäude.
Die neue Adresse lautet:
Eduard-Maurer Str. 9
Definition of accuracy and crosstalk
In order to define the accuracy of a multidimensional sensor, firstly two important terms should be distinguished:
The term "accuracy class", which appears in all data sheets and product descriptions, refers to the accuracy of the sensor.
In the document sensors: accuracy, there is a detailed explanation of this term and a summary of the most important criteria for the accuracy class:
For force and torque sensors, the following properties are used for classification into an accuracy class:
In the multi-axis and multi-component sensors occurs another effect, the so-called crosstalk. Here is an explanation from the document multi-component sensors:
The force or torque introduction into the measuring axis also results in an indication in the thereto perpendicular axes. This effect is referred to as crosstalk.
For three-axis force sensors and multi-component force sensors, the crosstalk when effects the nominal load is approx. 1% of the nominal load of the other axes.
Crosstalk is proportional to the amount of load. With increasing levers or larger moments the deformation of the sensor and the crosstalk is increasing.
The calibration takes place at the level of the front face of the sensor. In this level is also the origin of the multi-axis force sensor. In the actual application the fore application point (operating point) is usually not located in the origin of the multi-axis force sensor. This creates additional torques.
These additional torques must be considered during the selection of the measuring range to prevent overloading of the sensor.
If the torques should be calculated at the force application point, so is an additional arithmetic operation or alternatively a transformation of the calibration matrix necessary. The operation is described in the instructions ba-k6d.pdf. The software GSVmulti allows the input of the distance of the force application point from the coordinates origin.
The crosstalk in every operating point can be minimized via calibration in this point. The additional torques are taken into account during the calibration in the operating point and included in the error compensation.
The special option "Matrix Plus" allows the reduction of crosstalk in the operating point up to 0.2% or 0.5%. This option "Matrix Plus" includes:
The quadratic approach of the error compensation minimizes the crosstalk exactly in the operating point. At higher loads, the quadratic approach can lead to larger errors than error compensation with a linear approach.
The calibration with the load vector of the intended application also minimizes crosstalk for this load vector.
The selection of the respective procedure depends on the availability of calibration accessories and the individual requirements of your application.
A calibration with 100% nominal load is recommended for applications where the load vector is unknown. In order to minimize the crosstalk, an additional quadratic approach "matrix plus" may be applied with a mathematical optimization for the load cases 20%, 40%, 60% and 80% of the nominal load.
The value of the crosstalk is specified in each datasheet by multi axis and multi-component sensors separately from the accuracy class.
Even when using 1-axis force sensors, lateral forces and torques can be initiated. This also leads to a display of a measured value due to crosstalk. At constant lateral force or at a constant torque this leads to a shift of the zero point. This can be adjusted using the zeroing function or can be taken into account as an offset.
Accuracy and crosstalk are reported separately for 6-axis (force/torque) sensors. The accuracy class is typically given as 0.2%, the crosstalk as 1%. As a rule, the temperature-induced drift of 6-axis sensors leads to a reduction of the accuracy class to 0.2%. Reproducibility (standard uncertainty) and linearity are on the order of 1-axis force sensors with accuracy class 0.1%.
The special calibration methods minimize the crosstalk to the range of 0.2%...0.5%.