I am in need of a 3 axis force sensor having a resolution of 3mN. Does product K3D40 (10N or 2N) is suitable for such high resolution?
Thank you for your time,
the resolution is dependent of the amplifier. Normally you get Nominal Force divided by 10000 as resolution,
but with best amplifiers, you will get Nominal Force divided by 100000.
2N/10000 = 0.2mN at least.
Wer ist online?
Mitglieder in diesem Forum: 0 Mitglieder und 1 Gast